INTRODUCTION: As we saw at the beginning of this series, the 1m rule had a range of 1m! This may seem to be a statement of the obvious but when dealing with transducers, their range is of paramount importance. Range is crucially important for two main reasons.
Firstly the resolution of the signal conditioning is a proportion of the range it is set to which is fixed by the A/D converter’s bit range. For 12 bit A/D the range is divided into about 4000 steps. Secondly, and just as importantly, the transducer accuracy is specified by the manufacturer as a percentage of full range e.g. 0.05% FRO is common for pressure transducers. Thus range is the key element, not only in the specification of transducers and measuring systems, but also how these specifications work out in practice. These practical outcomes are best illustrated with some examples. To download the complete white paper, click on the PDF link below.
PLEASE NOTE: By clicking to download a white paper, you confirm that you have read and agree to the following provisions:
You are permitted to download, use and distribute copies of any white papers made available for download on this page, except that: You must only use and distribute the white papers in their entirety without amendment, deletion, text, graphics or other content.
The information contained in these documents is written by and is the opinion of GDS Instruments. White papers are for informational purposes and not reviewed by third parties.